News for VLBA/HSA/VLBI Proposers and Observers
February 1, 2017 — J. D. Romney
RDBE DDC Firmware Rewrite Completed, Testing Begun
The previous entry on this news page mentioned a rewrite of the DDC personality, with a goal of making the entire DDC capability compliant with standard firmware elements introduced since the initial version, and enabling real-time testing for better detection and analysis of faults. It is hoped that the new version will be relatively immune from the several types of rare delay and phase jumps seen in the initial firmware, and will allow any remaining problems to be isolated more effectively.
That 10-month rewrite project has now been completed, and testing of the new firmware has begun.
September 16, 2016 — J. D. Romney
RDBE Delay and Phase Jumps
Delay and phase jumps have been detected in observations made using both the DDC and PFB personalities of the RDBE. These occur only quite infrequently (as far as we are aware, at least), and characteristics of the jumps vary substantially, so it has not been possible to describe all the effects in detail, nor to recommend countermeasures beyond careful analysis of observational delays. General descriptions of the most common effects are sketched below.
1> Delay jumps upon resumption of an observation following interruption for short USNO operations at PT and MK: Any restart of an observation may require a reload of the appropriate FPGA firmware, and a resynchronization of the internal timekeeping may occur. A general rule for these cases is that return to delay and phase cannot be guaranteed on a schedule resumption, and comparison of delays before and after the interruption is necessary. Notification of the interruption has been included in the VLBA operators' logs for a long time, and was augmented recently to advise such a check.
2> DDC delay jumps: Such jumps have been seen sporadically for the last ~2 years, and have been the focus of a major investigation, and of numerous attempted corrections in the DDC firmware, all without complete success to this point. In some but not all cases, such jumps occur (if at all) on the second use of any given setup in the observation schedule, and do not recur subsequently. This effect, and the overall low rate of occurrence, both limit efficient analysis, but some test schemes have been developed to gather more data than available previously (at the cost of substantial observing time). Recent results suggest a basic jump quantum that is inversely proportional to the channel bandwidth: 256ns / BW[MHz] for 2 <= BW <= 16; jumps of this type are not known to occur for BW = 1 MHz nor BW >= 32 MHz. Some cases have been seen with small-integer multiples of that jump quantum. A major rewrite of the DDC FPGA firmware is currently under way, aimed both at preventing such jumps, and at facilitating tests to verify error-free performance.
3> PFB phase jumps: These have been rare enough to be thought not to occur any more, but at least one event was seen recently. The jump occurred only in the phase-cal results at one station, and was not seen in the actual interferometer phases.
September 10, 2015
RDBE_based VLBA Gains
VLBA pointing and gain measurements in the past in the past have been made based on power levels detected by the legacy Baseband Converters (BBCs). Since February 2014, power levels have also been measured in the new RDBE broadband system. The analysis which produces the gain values given in the vlba_gains.key has been run in parallel with both data sets. Until late August 2015, the vlba_gains.key derived from the BBCs was the one placed on the ftp site and used as the sources of the values distributed with exported data sets in the gain table. On September 2, 2015 we switched to using the gains derived from the RDBE data in the vlba_gains.key file. For more information, go .
January 2, 2015
2015B Call for Proposals Announced
July 7, 2014
2015A Call for Proposals Announced
January 5, 2014
2014B Call for Proposals Announced
The 2014B Call for Proposals was announced on January 5, 2014. Proposal deadline is February 3, 2014 at 17:00 Eastern Daylight Time (21:00 UTC).
July 8, 2013
2014A Call for Proposals Announced
The 2014A Call for Proposals was announced on July 8, 2013. Proposal deadline is August 1, 2013 at 17:00 Eastern Daylight Time (21:00 UTC).
January 3, 2013
2013B Call for Proposals Announced
The 2013B Call for Proposals was announced on January 3, 2013. Proposal deadline is February 1, 2013 at 17:00 Eastern Standard Time (22:00 UTC)..
December 31, 2011
VLBA flag problems (Nov 10 to Dec 22)
During the second week of November, upgrades were made at Pie Town in support of the upgraded C-band receiver. These changes have transferred monitor and control of some antenna electronics from the "legacy" station computer to the modern system which is based on EVLA software. This new system is working well, however the legacy system, which is still responsible for generating flags for the station, was perpetually flagging data based on monitor points that no longer exist. The Hancock antenna C-band system was upgraded in December with the same issue beginning with observations made on Dec 10.
Data observed after Nov 10 and correlated before Dec 7 will have a single erroneous flag for Pie Town encompassing the entire observation. Data observed between Dec 7 and Dec 22 won't have any flags for Pie Town or Hancock. Data observed after Dec 22 should be back to normal.
VLBA operations staff have regenerated flag data for the Nov 10 through Dec 22 period. Users wishing to restore proper flags can follow the instructions to download cal.vlba.gz and then follow instructions in the AIPS Cookbook. The section titled Automatic formatting of VLBA and VLBA-like log files will explain how to use VLOG and UVFLG to load flags from the downloaded file. You may want to make a backup (with TASAV) of the original FL table before deleting it (with EXTD) just in case you want to go back to the original flags.
Questions regarding the flagging problem and its corrections should be directed at the NRAO helpdesk.
August 29, 2011
The Hancock VLBA station was put into shutdown mode in the morning of August 27 in anticipation of high winds, heavy rains and power loss due to hurricane Irene. Commercial power was lost at approximately 3 am Sunday morning and has not yet been restored. The site techs were able to reach the station on Monday, August 29 and determined that there is no obvious damage to the antenna or structures. Numerous trees are down around the site and on the roads and power lines have been downed by fallen trees. It may take several days before power is restored, however, backup generators for critical systems (maser, cryogenics, etc.) are working so we anticipate a smooth return to operations once power is restored.
August 8, 2011
Error found in EOP corrections in AIPS
Versions of the AIPS task CLCOR released between Sept. 21, 2009 and Aug. 4, 2011 used the wrong sign on the station Y coordinate when calculating Earth Orientation Parameter (EOP) corrections. The main adverse impact will be on phase referencing projects for which sufficiently large EOP corrections were made that the difference in the relative correction between calibrator and target is significant. For more details click here.
July 6, 2011
Los Alamos antenna back in operation
In the evening of July 6, 2011 the danger posed by the Las Conchas fire had subsided sufficiently that the Los Alamos VLBA antenna was put back in operation. It has been taking part in VLBA observations since.
July 2, 2011
Las Conchas Fire near Los Alamos
The Las Conchas Fire that started on June 26, 2011, came within a few miles of the LA VLBA antenna. The LA station is out of danger, but the fire continues to burn and is not yet contained. The Los Alamos National Laboratory remains closed. NRAO therefore has no access to the antenna at this time, and LA cannot currently be included in VLBA observations.
June 30, 2011
Call for proposals for semester 2012A
The June 30, 2011 NRAO eNews contains the Call for Proposals for the GBT, EVLA, and VLBA/HSA/VLBI, along with general news for proposers.
January 7, 2011
Call for proposals for semester 2011B
The January 7, 2011 NRAO eNews contains the Call for Proposals for the GBT, EVLA, and VLBA/HSA/VLBI, along with general news for proposers.
September 15, 2010
Call for proposals for trimester 2011A
The September 15, 2010 NRAO eNews contains the Call for Proposals for the GBT, EVLA, and VLBA/HSA/VLBI, along with general news for proposers.
June 21, 2010
VLBA Pie Town azimuth track fixed
The Pie Town VLBA antenna was restored to full service on June 17, 2010, after antenna mechanics were able to effect a repair to the split in the rail. The damaged rail section was regrouted and the split rail was welded. It is believed that the split in the rail was caused by a poor quality weld of the rail to the mounting plate during the construction of PT. It appears that this weld cracked during cooling and then slowly propagated up the rail until the failure occurred.
June 1, 2010
VLBA Pie Town antenna down for repair of azimuth track
During a routine inspection of the Pie Town VLBA azimuth track on May 25, 2010, a split in the rail and grouting near bolt 20 was discovered. Until a repair is completed, the PT antenna will be unavailable for observing. Several options for the repair are under consideration, but PT will be out of service until at least July 1, 2010.
May 14, 2010
Call for proposals for trimester 2010C
The May 14, 2010 NRAO eNews contains the Call for Proposals for the GBT, EVLA, and VLBA/HSA/VLBI, along with general news for proposers.